This temperature transducer is intended for temperature measurement of loose, liquid and soft media in stationary and mobile control systems of different purpose.
The resistive temperature transducer is calibrated acc. to GOST 8.461.
This temperature transducer consists of a sensing element, which is located in a sharp-ended protection tube, an accessible shaft for temperature transducer manipulation and flexible cable for a connection with an external device.
|NSH||10М; 50М; 100М|
|Measuring temperature range,°С||minus 50 plus 100; minus 50 plus 160|
|Mounting part length, mm||100|
|Wiring diagram (circuit)||2-wire; 3-wire; 4-wire|
|Class||В or С|
|Thermal response time, no more than, s||60|
|Fastening||Mounting to a protection sleeve|
|Protective fitting, mm||∅4|
|Recalibration interval, years||2|
A connecting with external devices is made with the help of the proper section of a flexible cable and proper amount of wires.
|Design||∅D, mm||Lраб, mm||Fastening||NSH||Class||Circuit||tраб,°С|
|ТСМ-Н 8.0.02.00.7.1.2||4||100||50 М||0,00428||В||2-wire||-50...+160|