This temperature transducer is applied for outdoor temperature measurement in systems of control and automation regulation of different purpose
The temperature transducer should be used with a protective sleeve.
The temperature transducer consists of a sensing element, which is located in a protection tube ∅6mm, and of a cable with the proper amount of wires and proper section.
The isolation of joining wire tips is removed. Dealing with 4-wire circuit, pairs of wires have different length in accordance with the inner connection.
|Measuring temperature range,°С||from 0 to plus 160 from minus 50 to plus 180|
|Mounting part length, mm||45; 60; 80; 100; 120; 160; 200; 250|
|Wiring diagram (circuit)||2-wire; 4-wire;|
|Class||A or В (СТБ EN 60751), AА|
|Fastening||Mounting to a protection tube|
|Protective fitting, mm||∅6|
minus 50-180°С recalibration interval 4 years.
The temperature transducer is mounted to a pipeline with the help of protection sleeve and weldolets.
Protective sleeve GZ5/8-xx-G1/4 and weldolets BP3 are used for mounting of the temperature transducer with length 45 and 60 mm.
Protective sleeve GZ2/8 and weldolets BP1(or BP2) are used for mounting of temperature transducer with length 45 and 60 mm.
Connecting with an external device is made with the help of the proper section of a flexible cable and the amount of wires corresponding to the temperature transducer circuit.
|Design||∅D, mm||Lраб, mm||Fastening||NSH||Class||Circuit||tраб,°С||Cable, m|
|ТСП-Н 1.1.21.00.3.3.0/1.5||6||45||In a sleeve||Pt100||А||4-wire||0...+160||1,5|
|ТСП-Н 1.1.02.00.4.1.1/0.5||6||100||In a sleeve||Pt100||В||2-wire||-50...+180||0.5|
|ТСП-Н 1.1.00.00.7.1.0/2||6||60||In a sleeve||Pt500||В||2-wire||0...+160||2|
|ТСП-Н 1.1.01.00.3.2.1/1||6||80||In a sleeve||Pt100||А||3-wire||-50...+180||1|
|ТСП-Н 1.1.21.00.3.3.1/1||6||45||In a sleeve||Pt100||А||4-wire||-50..+180||1|