ТThis temperature transducer is applied for outdoor temperature measurement in systems of control and automation regulation of different purpose.
The construction of the temperature transducer casing makes it possible to mount it immediately to the wall or other surface with screws, woodscrews and fastening meanings.
The leaden seal is fixed through a lid opening.
The temperature transducer represents a phenoplastic casing with an unscrewing lid, a protection tube of stainless steel and a gland entry for a cable. A sensing element is settled in a protection sleeve.
Electrical connection of the sensing element with screws for an external cableconnection is made corresponding to 2-wire, 3-wire or 4-wire circuit (the temperature transducers corresponding to class A with 2-wire circuit are not made)
Connecting with external devices is made with the help of the appropriate section cable and the amount of wires recommended in the cicuit.
The cable connects to the circuit with the help of screws and nuts.
|Measuring temperature range,°С||from minus 50 to plus 180|
|Mounting part length, mm||60; 80|
|Wiring diagram (circuit)||2-wire; 3-wire; 4-wire; 2-x 2-wire|
|Class||АА, А, В (GOST 6651)|
|Thermal response time, no more than, s||30|
|Protective fitting, mm||∅6; ∅8|
|Recalibration interval, years||4|
Connecting Diagram of the sensing Elements
The Temperature Transducer Design Examples
|Design||∅D, mm||Lраб, mm||Fastening||NSH||Class||Circuit||tраб,°С|
|ТСП-Н 2.2.00.00.7.2.1||8||60||in-situ||Pt500||В||3-wire||-50.. +180|