This temperature transducer is applied for temperature measurement in systems of control and automatic regulation of different purpose.
Тhe temperature transducer represents a phenoplastic casing with an unscrewing lid, a protection tube of stainless steel and a gland entry for a cable. A sensing element is settled in a protection sleeve.
Electrical connection of the sensing element with screw for an external cable connection is made corresponding to 2-wire, 3-wire or 4-wire circuit.
Connection with external devices is made with the help of the appropriate section cable and the amount of wires recommended in the circuit.
The cable connects to the circuit with the help of screws and nuts.
|Measuring temperature range,°С||from 0 to plus 160 from minus 50 to plus 180|
|Mounting part length, mm||60...3150|
|Wiring diagram (circuit)||2-wire; 3-wire 4-wire; 2-х 2-wire|
|Class||А or В (СТБ EN 60751) or, AА|
|Thermal response time,s||30|
|Protective fitting,mm||∅6 ∅8|
from minus 50 to 180 recalibration interval 4 years for other - recalibration interval 2 years.
Connecting Diagram of the Sensing Elements
The temperature transducer is mounted to a pipeline with the help of protective sleeves and weldolets or immediately with the help of the appropriated fastening elements of the temperature transducer dependently on a performance of fastening elements.
The leaden seal is fixed through a lid opening, or through a seal opening in a fitting.
Fastening Elements Design Examples
Installation in a protective sleeve
It’s recommended to mount the temperature transducer without any installation elements to a pipeline with the help of protective sleeve of design 2 and with the help of weldolet BP1 (or BP2).
It’s recommended to mount the temperature transducer with a mobile fitting to a pipeline with the help of protective sleeve of design 1 and weldolet BP1 or BP2 or it’s also possible with the help of protective sleeve of design 4 for immediate installation at a pipeline.
The temperature transducer with mobile nut and fixed fitting are applied for immediate mounting to a pipeline with a client’s fastening element.
|Design||∅D, mm||Lраб, mm||Fastening||NSH||Class||Circuit||tраб,°С|
|ТСП-Н 3.1.00.02.4.1.0||6||60||Mobile fitting М20х1,5||Рt100||В||2-wire||0...+160|
|ТСП-Н 3.1.01.00.3.3.1||6||80||Protective sleeve||Рt100||А||4-wire||-50...+180|
|ТСП-Н 3.2.02.15.7.1.0||8||100||Fixed fitting М20х1,5||Pt500||В||2-wire||0...+160|
|ТСП-Н 3.2.03.15.6.3.1||8||120||Fixed fitting М20х1,5||Pt500||А||4-wire||-50...+180|
|ТСП-Н 3.2.04.00.7.3.1||8||160||Protective sleeve||Pt500||В||4-wire||-50...+180|