This temperature transducer is applied for temperature measurement in systems of control and automatic regulation of different purpose with a short response time.
The temperature transducer should be used with a protective sleeve.
The temperature transducer consists of a sensing element, which is located in a protection tube,∅4, also of a threaded bushing, terminal block, corps of aluminum and a lid.
A lid is mounted to the corps through a rubber washer and is fixed with two screws. Thus pressure tight joint is achieved.
A threaded bushing has an oblique cut to fix a temperature transducer in a protection sleeve to prevent access to the circuit without any sanction without breaking the leaden seal.
The temperature transducer is mounted to a pipeline with the help of protective sleeves of design 3 and weldolets BP1 or BP2.
The gap clearance between a protective sleeve and a threaded bushing slot is 2mm.
A leaden seal is fixed through a sleeve screw, a corps opening and an opening of two screws which tighten a lid of the temperature transducer.
|Measuring temperature range,°С||from 0 to plus 160 from minus 50 to plus 180|
|Mounting part length, mm||60; 80; 100; 120; 160; 200; 250|
|Wiring diagram (circuit)||2-wire; 4-wire;|
|Class||А or В (CTБ EN 60751)or АА|
|Thermal response time, no more than, s||6|
|Fastening||Mounting to a protective sleeve|
|Protective fitting, mm||∅4|
minus 50 - plus 180°С recalibration interval 4 years.
Connecting with an external device is made with the help of the proper section cable and the amount of wires corresponding to the temperature transducer circuit.
The cable is connected with the circuit through a terminal block with the help of nuts and pucks.
|Design||∅D, mm||Lраб, mm||Fastening||NSH||Class||Circuit||tраб,°С|
|ТСП-Н 5.0.00.00.4.1.0||4||60||in a sleeve||Рt100||В||2-wire||0...+160|
|ТСП-Н 5.0.01.00.3.3.1||4||80||in a sleeve||Рt100||А||4-wire||-50...+180|
|ТСП-Н 5.0.02.00.7.1.0||4||100||in a sleeve||Pt500||В||2-wire||0...+160|
|ТСП-Н 5.0.03.00.6.3.1||4||120||in a sleeve||Pt500||А||4-wire||-50...+180|
|ТСП-Н 5.0.04.00.6.3.0||4||160||in a sleeve||Pt500||А||4-wire||0..+160|