The resistive temperature transducer is intended for temperature measurement of loose, liquid and gaseous media in stationary and mobile control systems of different purpose.
This temperature transducer should be used with a protection sleeve.
The temperature transducer consists of a sensing element in a protection tube c6 and a terminal block.
Constructively this temperature transducer is intended for usage in a proective sleeve with a hermetic corps of aluminum. fastening is made with the help of spring-loaded screws, which tightly pressthe temperature transducer to a protective sleeve base.
The temperature transducer is mounted to a pipeline with the help of protective sleeve GZ8-X and weldolets BP1(or BP2) or without it.
The temperature transducer should be installed in a protective sleeve, then the connection with an external device and covering a lid and fixing it with a screw should be made, creating tight joint.
A leaden seal is fixed through a fixing lid screw and it helps to prevent from an access to the circuit.
|Measuring temperature range,°С||from 0 to plus 160 from minus 50 to plus 180|
|Mounting part length, mm||160; 200; 250; 320; 400|
|Wiring diagram (circuit)||4-wire; 2-wire;|
|Class||А or В (СТБ EN 60751), AА|
|Fastening||Mounting to a protection sleeve|
|Protective fitting, mm||∅6|
minus 50 - plus 180°С recalibration interval 4 years.
Connection with an external device is made with the help of the proper section cable with the amount of wires and corresponding to the temperature transducer circuit.
|Design||∅D, mm||Lраб, mm||Fastening||NSH||Class||Circuit||tраб,°С|