The set is intended for measurement of temperature and temperature difference in systems of control and automatic regulation of heat quantity.
The set is used for an installation at pipelines to the point of DN 25.
The set is selected according to the method MP VT 047-2002 in three points of the resistive temperature transducer ТСП-Н.
The temperature transducer mounted to a delivery pipeline is signed by letter “Г”, and mounted to a return pipeline by letter “X”.
They are installed immediately in valves, T-couplers, etc. through a transitive nut.
When they are installed immediately in a ball valve, it’s became unnecessary to lighten the systems while mounting and to change a temperature transducer after the recalibration interval has been finished.
|Measuring temperature range,°С||from 0 to plus 160|
|Mounting part length, mm||27,5|
|Wiring diagram (circuit)||2-wire; 4-wire;|
|Class||В (СТБ EN 60751)|
|Fastening||Mounting to a protection sleeve|
|Protective fitting, mm||∅3,7|
|Design||∅D, mm||Lраб, mm||Fastening||NSX||Class||Circuit||Δt,°С||Cable, m|
|КТСП-Н 6.0.00.11.7.1.3/1||4||27,5||Fitting М10x1,0||Pt500||В||2-wire||3...150||1|
|КТСП-Н 6.0.00.11.4.1.3/1.5||4||27,5||Fitting М10x1,0||Pt100||В||2-wire||3...150||1,5|