The set is intended for measurement of temperature and temperature difference in systems of control and automatic regulation of heat quantity.
The temperature transducers from the set should be used with protective sleeves.
The set is selected according to the method MP VT 047-2002 in three points of the resistive temperature transducer ТСП-Н.
The temperature transducer installed at a delivery pipeline is signed by a red screw of a gland entry and installed at a return pipeline by a blue screw.
The temperature transducer is mounted to a pipeline with the help of protective sleeves of design 2 and weldolets БП1 (or БП2).
The cable is connected with the circuit through a terminal block with the help of nuts and pucks.
|Measuring temperature range,°С||from 0 to plus 160|
|Mounting part length, mm||60; 80; 100; 120; 160; 200; 250|
|Wiring diagram (circuit)||2-wire; 4-wire;|
|Class||А or В (СТБ EN 60751)|
|Fastening||Mounting to a sleeve|
|Protective fitting, mm||∅6|
|Design||∅D, mm||Lраб, mm||Fastening||NSX||Class||Circuit||Δt,°С|
|КТСП-Н 5.1.02.00.3.3.3||6||60||In a sleeve||Pt100||A||4-wire||3...150|
|КТСП-Н 5.1.03.00.4.1.2||6||80||In a sleeve||Pt100||В||2-wire||2...150|
|КТСП-Н 5.1.04.00.3.3.3||6||100||In a sleeve||Pt100||A||4-wire||3...150|
|КТСП-Н 5.1.04.00.7.1.2||6||100||In a sleeve||Pt500||В||2-wire||2...150|
|КТСП-Н 5.1.05.00.6.1.3||6||120||In a sleeve||Pt500||A||2-wire||3...150|